I'm using TINA-TI and it's all good apart from I can't get the oscilloscope to sync properly to the waveform and display it without scrolling along. On a real scope you adjust the trigger level to stop the wave scrolling left to right, but nothing I try with the TINA scope seems to make any difference. Anybody got any ideas or experienced this problem?
Under the trigger section, first set the mode to Normal instead of Auto. Then set the source to the signal you are interested in. You may want to use the little probe down towards the bottom to click on the circuit node of interest. Then you may have to hit the Run button again. Good luck!
- This old topic is closed. If you want to reopen this topic, contact a moderator using the "Report Post" button.