Intelligent Curve Tracer 3.0 release
This version use USB or RS232 interface communication with PC.
It can measure BJT's Ic->Hfe curve
Can measure Vce-Ic curve
measure some semiconductor transmission curve,such as Vbe-Ic / Vgs-Id
also FETís Vds-Id curve measure,Even can study Vgs & Rds relationship
Upper is six IRF630 Vds-Id curves,their Vgs is 5V 6V 7V 8V 9V 10V.
The Vds-Id curve only show Vds=0~1V range,so you can use systemís calculate function get the slope value,
The window in the central show six curveís slope, Measurement unit is V/mA, multiplicate 1000 is Rds,
So under Vgs=10V,Rds is 0.000334649870637804*1000=0.334ohm
Some diode V->I curve
fllower is TL431 in all current range V-I curve
if you plug TL431 into socket B and E,and change the measure condition,you can see less current VI curve. The following figure , Y-axis is only 0~0.38mA
the Intelligent Curve Tracer is more powerful,It can measure 3 terminal voltage regulatorís Vi-Vo / Io->Vo Curve
Intelligent Curve Tracer Principle
Before talking about my Intelligent Curve Tracer principle,I talk about the general measure curve principle by MCU first.
The general measure diagram as in Figure
Mcu control Channel B/C DAC and OCL power amplifier to driving the DTU, And measure four voltage(VBH/VB/VC/VCH) by ADC, and then calculate the Ib/Ic/Vce/Vbe and then draw the Vbe-Ic/Vce-Ic/Ic-Hfe curve.
Because of the need for the measure of P and N type semiconductor,so all the voltage are bipolar.
But considering the component selection, Most of the DAC and ADC can only output or measure unipolar voltage,bipolar ADC price will be very expensive.
Considering the cost of it, it uses unipolar ADC/DAC a lot cheaper, choice scope is much extensive.
So if we use unipolar ADC/DAC to design the curve tracer,we need some additional circuit:
1.four amplifier,It’s gain is –1, purposes to change negative voltage to positive voltage,so that the ADC can handle.
2.two amplifier to offset DAC to bipolar voltage;
3.Need biporlar power supply, but the DTU’s vce voltage range is only half of the total supply voltage;
4.Need 8 channle ADC to measure vbh1/vbh2/vb1/vb2/vch1/vch2/vc1/vc2;
My intelligent Curve Tracer in the above principle to make an improved, Using the" three voltage amplifier to drive DTU, the DTU emitter isn’t connect to ground, but from a voltage source,.
the following diagram principle:
The emitter voltage source(Channel E Power Amp) is very critical, completely solve the shortcomings of the above,.
Measure NPN BJT,We set Channel E Power Amp to output 2.5V, and set Channel B/C Channel power amplifier to output more then 2.5V,such as B channel is 5V, C channel is 20V,so the NPN can work. We measure the Ve/Vbh/Vb/Vc/Vch then calculate the Vbe/Vce/Ib/Ic/Hfe and draw the curve.
If measure PNP BJT,We set Channel E to output 40V, and set Channel B/C Channle to output less then 40V,so PNP can work. We also measure the Ve/Vbh/Vb/Vc/Vch then calculate the Vbe/Vce/Ib/Ic/Hfe and draw the curve.
If measure N type J-FET,We connect JFET’s source to E, grid to B, Drain to C. and set E channel output 10V,and set B Channel is from 2.5V to 10V, and set C Channel output from 10 to 40V. This means that the JFET’s Vgs is –7.5 to 0V,and Vds is from 0 to 30V. This is the JFET normal operating voltage range.
You look back above example, All the measure voltage are unbipolar,so DAC and ADC are just use a unipolar model device.
And the solution only need a 40V unbipolar power supply, and the DTU Vce range can be from 0 to 40V.
1.three channel power amplifiers
2.power supply and protect circuit
3.USB and RS232 interface circuits
4.MCU & ADC & DAC & Control circuit
Very good job!!!
What are the max Vce/Vds and Ic/Id that can work?
I will measure also the newer Sic Power Jfet, the SIT and the old V-fet. :D:D:D
Let me know.:cool:
I want buy it as finite and complete product.
Ebay or Paypal is possible?
Good work! It would be interesting to have additional interface connections to allow for a higher voltage driver circuit to use for testing vacuum tubes. It might be worthwhile to be able to generate stepped control for the screen grid also. Some curves that people usually do not see, like transconductance versus screen voltage could also be possible. It might help people know how much to vary the screen voltage of one tube to balance in a pair with another that does not match. Some might consider advanced options like sweeping the curves for one tube, have them in memory, then additionally stepping the screen voltage on a second tube to find the closest match. The common practices of matching gain at one bias value with a tube tester, and adjusting amplifier bias for equal idle current works poorly. A great improvement in matching and in selection of settings to compensate for mismatch is possible with more advanced testing.
If you current is 2A ,and current sampling resistor select 4.5ohm,then
max Vce is only 36-2*4.5=27V.
But if you measure JFET,the absolute(Vgs) + Vds + Vrc must less 36V,
Example: if you set Vgs is -5V, current is 2A,and sampling resistor is 4.5ohm
then max Vds is 36-5-2*4.5=22V.
The device sell link
Wholesale Intelligent Curve Tracer
On ebay,I has been registered,but can‘t sell,because ebay can't let me to 'go through an additional verification step' (需通过eBay跨国交易手工认证).
And aliexpress and ebay is 'competitive relation'.So aliexpress can't use palpay after 2011.08.
but 2 years ago,I had development some tube curve tracer.
Two anode/cathode/grid voltage Generator and high side current measure board
measure 6N13(Chinese tube)
Set grid voltage as -90/-85/-80/-75/-70/-65/-60/-55/-50/-45/-40/-35/-30/-25/-20/-15/-10/-5V and measure the plate voltage->Plate current curve.
The 6N13 transmission curve in different anode voltage,the anode voltage is 25/50/75/100/125/150/175V
the 6N12(double tube) transmission curve,Objective to understand the degree of match
but the tube curve precision not enough at low grid-cathode voltage,
you can compare 6N12 transmission curve and 6N13 transmission curve,
the 6N13 is more smooth,becasue 6N13 grid-cathode voltage is more high.
Tube curve tracer had termination.
Are there any additional taxes in China, when sell to outside or the shipping cost to Europe is to high?
I see at Taobao a price of 560 CNY...almost 88$ US and at Wholesale a price of 229$US
You can find the cost to China from the United States send 3Kg of cargo. Estimated from China to the United States is nearly.
The 'Trading platform fee' on Ebay of China is 6~7%,Aliexpress nearly.
The 'Cash withdrawal fee' of Palpay of China is $35,Aliexpress nearly.
May be have some Customs Tariff.
Can you sell the kit without the box ? it might be cheaper for DIYAUDIO members to buy( particularly from US ) avoiding excessive shipping costs etc?
Locky, my curve tracer arrived last day.
I have connected it to pc with serial interface RS232. Then, in mený configruration parameter, i have selected "Autodetect" and the answer was "DEVICE NOT FOUND".
I have checked the switch was in RS232 mode. My power line is 220 AC, I am not able to view if power is "ON" because no led installed, but the wires from AC plus was ok. So I i tell you "because the device not respond?":confused:
I not get program association for the .PRE extension file. My O.S. is Windows XP
Thanks much, Francesco.
|All times are GMT. The time now is 09:01 AM.|
vBulletin Optimisation provided by vB Optimise (Pro) - vBulletin Mods & Addons Copyright © 2016 DragonByte Technologies Ltd.
Copyright ©1999-2016 diyAudio