|Tubes / Valves All about our sweet vacuum tubes :) Threads about Musical Instrument Amps of all kinds should be in the Instruments & Amps forum|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|10th April 2013, 04:11 PM||#1|
Join Date: Dec 2012
Stark 9-66 tube tester - repair help
I have an old Stark 9-66 tube tester (Hickok 533A clone) that I wish to repair. I own it since 2 days so I know a little about it. I did some research on the net and found very good info on this website -> pacifictv.ca, here's the schematic http://pacifictv.ca/schematics/stark966schematic.pdf
The problem is that I can not properly adjust the "line level", I hold the P7 button while I turn the "Line adjust" knob, the needle is going right and left but far below the expected center spot.
I would really appreciate if anybody have some suggestions that could help me.
I found the problem, now 100% working..
Last edited by sparchead; 11th April 2013 at 03:18 PM.
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|calibrate stark 8-77?||jonnymyaudio||Tubes / Valves||0||20th January 2013 01:29 AM|
|Stark 9-66 tester repair||cancon||Tubes / Valves||2||16th June 2011 05:44 AM|
|stark and ciare CH250's in toronto||smak||Swap Meet||8||23rd March 2011 07:24 PM|
|Celestion Ditton 66 midrang needs repair||cbriere||Multi-Way||0||10th December 2007 07:35 PM|
|Old Stark meter - how to???||JesseG||Parts||17||9th May 2006 09:38 PM|
|New To Site?||Need Help?|