|Tubes / Valves All about our sweet vacuum tubes :) Threads about Musical Instrument Amps of all kinds should be in the Instruments & Amps forum|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|12th May 2011, 08:20 PM||#1|
Join Date: Aug 2010
Hickok 539A issue
Hello everyone,I recently picked up a Hickok 539A and it seems to work well except for the following issue:
When the suppressor switch is in position two:
- I lose bias voltage when the shorts selector is in "tube test".
- The shorts light illuminates in all positions except one.
- The following voltages are present during a tube test:
(Setup for a 6V6 using pin 8 as a ground)
Pin 1: 0VDC,3.3VAC
Pin 2: 0VDC,3.3VAC
Pin 3: 197VDC,24.8VAC
Pin 4: 137VDC,37.2VAC
Pin 5: 0VDC,0VAC
Pin 6: 0.35VDC,7.86VAC
Pin 7: 0VDC,3.26VAC
- The following voltages are present during a shorts test:
Pin 1: 0VDC,0VAC
Pin 2: 0VDC,3.2VAC
Pin 3: 0VDC,0.77VAC
Pin 4: 0VDC,0.77VAC
Pin 5: 0VDC,0VAC
Pin 6: 0VDC,10VAC
Pin 7: 0VDC,3.2VAC
When the cathode selector is in position two:
- I lose bias voltage
- The shorts light does not illuminate in any position.
I've tried cleaning the sockets with some contact cleaner to no avail. Changing any of the other selector switches has no effect on the above issues. The 83 and 5Y3 have been swapped in from a known working unit.
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|Help with Hickok 539A tube tester repair..||Mort Caldwell||Tubes / Valves||14||5th November 2013 12:32 PM|
|Test settings for Hickok 539A for 183/483 tube?||Rick R||Tubes / Valves||1||3rd April 2011 06:04 PM|
|Anyone Have a Hickok 800A||skidave||Tubes / Valves||6||22nd December 2008 03:30 PM|
|Hickok 533A help||cwujek||Tubes / Valves||4||20th July 2007 01:09 PM|
|Hickok 539A settings||John S.||Tubes / Valves||4||27th December 2001 04:16 AM|
|New To Site?||Need Help?|