
Home  Forums  Rules  Articles  diyAudio Store  Gallery  Blogs  Register  Donations  FAQ  Calendar  Search  Today's Posts  Mark Forums Read  Search 
Tubes / Valves All about our sweet vacuum tubes :) Threads about Musical Instrument Amps of all kinds should be in the Instruments & Amps forum 

Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving 

Thread Tools  Search this Thread 
28th October 2010, 08:18 PM  #21 
diyAudio Member
Join Date: Mar 2003
Location: Haarlem, the Netherlands

Hi DF96,
I use the parallel connection of a 1 Mohm and a 47 kohm resistor as a "calibrated noise source" and assume that the valve noise can be represented with sufficient accuracy with just an equivalent input noise voltage source. This is obviously not true for the Amperex device with 1 uA grid current, which is why I didn't bother to measure its noise. So what I measure is: Noise of the anode bias resistor, amplifier and laptop with the valve supplies switched off Noise of the valve plus the anode bias resistor, amplifier and laptop with the grid shorted to ground Noise of 44890 ohm (47 kohm//1 Mohm) plus the valve plus the anode bias resistor, amplifier and laptop, the 44890 ohm being connected between grid and ground The thermal noise of 44890 ohm is known, so with a bit of straightforward mathematics I can calculate the equivalent input noise voltage of the valve from these results. The impression I get from my measurement results is that white noise drops with anode current, whereas the 1/f noise has a minimum somewhere around 2.5 mA anode current. That white noise drops monotonically while 1/f has a minimum is as expected from theory. But I'll do some more spot noise measurements to see whether I'm right. Best regards, Marcel van de Gevel 
28th October 2010, 09:00 PM  #22 
diyAudio Member
Join Date: May 2007

Thanks for the clarification. I will have another think about your results.

28th October 2010, 10:19 PM  #23 
diyAudio Member
Join Date: May 2007

Assuming a typical triodeconnected gm of 2.5mA/V, Req should be around 1K which will give noise of 4nV/sqrt(Hz). Your figures are about 6dB above this, although flicker noise makes a contribution at low frequencies. It might be possible to estimate the 1/f corner frequency, somewhere around 1kHz.
Interesting results! 
29th October 2010, 10:50 PM  #24 
diyAudio Member
Join Date: Mar 2003
Location: Haarlem, the Netherlands

Additional spot noise values, second Trigon device (actually the old recordings processed differently):
100 Hz to 150 Hz: (193.5 uA: 22.53 nV/sqrt(Hz)) (298 uA: 20.86 nV/sqrt(Hz)) 907 uA: 22.02 nV/sqrt(Hz) 2073 uA: 33.16 nV/sqrt(Hz) 3960 uA: 43.81 nV/sqrt(Hz) 10 kHz to 12 kHz: 907 uA: 9.75 nV/sqrt(Hz) 2073 uA: 5.77 nV/sqrt(Hz) 3960 uA: 5.68 nV/sqrt(Hz) In retrospect I don't trust the measurements at 193.5 uA and 298 uA: I heard sounds that didn't sound like noise, so maybe I was measuring mains voltage fluctuations and/or microphony. Also the 298 uA 100 Hz to 150 Hz value makes no sense when you compare it to the 20 Hz to 20 kHz value. Ignoring these values, the general trend is that between 907 uA and 3960 uA, white noise gets less with increasing current while 1/f noise increases. Apparently the 1/f noise minimum for Trigon EF86 number 2 occurs at an anode current of less than 907 uA. For the Amperex devices the lowfrequency noise increased very rapidly below 1 mA. I didn't measure that, but I could hear it very clearly. 
29th December 2010, 11:00 AM  #25 
diyAudio Member
Join Date: Mar 2003
Location: Haarlem, the Netherlands

Hi,
I've remeasured the triodeconnected Trigon EF86 valve at currents below 1 mA and I've reprocessed the old recordings for 2 mA and 4 mA. The latest results are: Flat, 20 Hz to 20 kHz: 199 uA: 16.45 nV/sqrt(Hz) 297 uA: 13.87 nV/sqrt(Hz) 891.5 uA: 9.045 nV/sqrt(Hz) 2073 uA: 7.384 nV/sqrt(Hz) 3960 uA: 8.541 nV/sqrt(Hz) Aweighted, 20 Hz to 20 kHz: 199 uA: 15.73 nV/sqrt(Hz) 297 uA: 13.11 nV/sqrt(Hz) 891.5 uA: 8.438 nV/sqrt(Hz) 2073 uA: 6.482 nV/sqrt(Hz) 3960 uA: 6.357 nV/sqrt(Hz) 100 Hz to 300 Hz: 199 uA: 19.9 nV/sqrt(Hz) 297 uA: 20.02 nV/sqrt(Hz) 891.5 uA: 20.8 nV/sqrt(Hz) 2073 uA: 26.27 nV/sqrt(Hz) 3960 uA: 38.03 nV/sqrt(Hz) 1 kHz to 1.2 kHz: 199 uA: 16.32 nV/sqrt(Hz) 297 uA: 14.56 nV/sqrt(Hz) 891.5 uA: 10.01 nV/sqrt(Hz) 2073 uA: 9.062 nV/sqrt(Hz) 3960 uA: 8.871 nV/sqrt(Hz) 10 kHz to 12 kHz: 199 uA: 16.21 nV/sqrt(Hz) 297 uA: 12.89 nV/sqrt(Hz) 891.5 uA: 8.205 nV/sqrt(Hz) 2073 uA: 5.798 nV/sqrt(Hz) 3960 uA: 5.597 nV/sqrt(Hz) Needless to say, I haven't a clue how accurate these measurements are. At least you clearly see the trends: white noise gets less with increasing bias current while excess noise increases with increasing current. The total noise around 200 Hz stays more or less constant up to about 1 mA and then increases. The optimal current for audio use is in the range of 2 mA to 4 mA, presumably on the lower side of this range for RIAA amplifiers as they have more gain for lower frequencies. The trend for 1/f noise below 1 mA was very different on the crappy Amperex devices, there it increased enormously below a certain current. Presumably this was the island effect that Van der Ziel wrote about. My measurement procedure is now: Switch off all equipment that you don't need, preferably by unplugging it, to minimise interference Record the noise with the valve switched off Record the noise at various bias current settings with grid grounded Record the noise at various bias current settings with grid connected to a known resistor that is used as a predictable noise source (44890 ohm for the old measurements, 20 kohm +/ 1 % for the newer measurements) Use a comb filter in GoldWave to suppress 50 Hz mains hum and its harmonics (wave(n)wave(n+882) at 44.1 kHz sample rate) Use a steep 20 Hz highpass to remove any subsonics related to mains voltage variations (20 Hz, steepness=20 in GoldWave, which actually means a 40th order filter) Listen very carefully to each recording and select a part without any noises due to microphony and thermal expansion, the neighbour's GSM telephone, ticks or anything else that is not noise Maximise the level and write down how much GoldWave has increased the level Measure the RMS level with GoldWave's "Volume Match" function. Volume Match determines the RMS value but neglects anything below a certain threshold, hence the need to first maximise the level to get an accurate result. For the spot noise measurements, bandpass filter the signal, maximise the level and write down how much GoldWave has increased it, cut off a small part from the beginning because the bandpass filter needs some time to settle, use Volume Match again to measure the RMS value. Don't make the measurement bandwidth too narrow if you have only short sound fragments left. Use some straightforward mathematics and a spreadsheet to correct for the measurement amplifier noise and to determine the valve noise by comparing it to the known noise from the resistor (assuming that the valve has only equivalent input voltage noise, no input current noise). Best regards, Marcel Last edited by MarcelvdG; 29th December 2010 at 11:07 AM. 
29th December 2010, 02:50 PM  #26 
diyAudio Member
Join Date: Sep 2003
Location: Alps:Tube amp designs over 150W, SMPS guru.


30th December 2010, 06:19 PM  #27 
diyAudio Member
Join Date: Mar 2003
Location: Haarlem, the Netherlands

Hello Richy,
Thanks for the interesting link! I'm surprised that Standard Telephones and Cables Pty Ltd blaims the leakage of the mica insulators for the spread on the noise, particularly at a source impedance as low as 100 kohm. According to Van der Ziel current noise due to leakage is usually negligible except when unusually high impedances are used, like in electrometer circuits (or condenser microphone amplifiers). Besides, he attributes the leakage to electrons from the cathode being picked up by the grid, electrons being emitted from the grid because of high grid temperature or light falling on the grid, and to collision ionisation due to traces of gas. So basically anything you can think of except the mica insulators. Best regards, Marcel 
31st December 2010, 10:15 AM  #28 
diyAudio Member
Join Date: Mar 2003
Location: Haarlem, the Netherlands

By the way, according to Van der Ziel collision ionisation due to small traces of gas can also increase the white voltage noise enormously (which would be visible at low source impedances), even at small grid currents.
"The positive ions, however, moving through the potential minimum with low speed, reduce the space charge and thus give rise to a large temporary increase in anode current. The original current fluctuation thus becomes amplified by a factor mu_0 which is rather large (mu_0=10 or more)." He shows a measured graph taken from B. J. Thompson and D. O. North, RCA review, vol. 5, nr. 371, 1941. For the specific valve that they measured a gas pressure giving 1 uA of grid current gave an extra noise of sqrt(2q*4 mA)~=35.8 pA/sqrt(Hz) at the anode. At a transconductance of (say) 2.5 mA/V that would correspond to 14.3 nV/sqrt(Hz) equivalent input noise voltage. 
31st December 2010, 10:41 AM  #29 
diyAudio Member
Join Date: Aug 2006
Location: Berkshire UK

Lo noise valves/tubes
Hi all,
Another couple to chuck into the equation are the Brimar 6BR7 and 6BS7, Which I see someone has already commented on the 6BR7. I believe the 6BS7 was specifically for instrumentation as it has a g1 top cap. John Caswell Last edited by John Caswell; 31st December 2010 at 10:44 AM. Reason: Further comments 
13th January 2013, 09:45 PM  #30 
diyAudio Member
Join Date: Apr 2007

Hello MarcelvdG,
could do you detail your noise measurements setup, please ? Thanks ! 
Thread Tools  Search this Thread 


Similar Threads  
Thread  Thread Starter  Forum  Replies  Last Post 
Low space / Low cost B20  Trooper46  Full Range  5  13th September 2008 06:01 PM 
Extremely low power and low loss transformers  percy  Tubes / Valves  12  23rd February 2005 11:30 AM 
Best lownoise lowdistortion buffer?  borges  Solid State  7  9th December 2003 01:40 PM 
New To Site?  Need Help? 