Oscilloscope Probes for Tube Work
I have just purchased an old Tektronix 465B and need to get a suitable set of probes for working with tube gear.
In Building Valve Amplifiers, Morgan Jones writes: "Typical X10 probes can only withstand < 200V, and the HT at the power amplifier output stage is likely to be >300V. In addition to your X10 probes, you will also need an X100 probe that is rated for high voltages."
Having picked up some probes rated "<600VDC + Peak AC", I thought I was pretty well covered for anything with a B+ of 500V or less (giving myself a 20% margin of error). Am I missing something?
I think you'll be fine, if the probes actually have a rating on them. The cheap ones I have here have nothing specified (didn't stop me from using them of 450V)!
Assume the 465 has the usual 400V input rating itself...
Re: Oscilloscope Probes for Tube Work
|All times are GMT. The time now is 05:44 PM.|
vBulletin Optimisation provided by vB Optimise (Pro) - vBulletin Mods & Addons Copyright © 2017 DragonByte Technologies Ltd.
Copyright ©1999-2017 diyAudio