Making a milliAmp meter, become a Volt meter - diyAudio
 Making a milliAmp meter, become a Volt meter
 User Name Stay logged in? Password
 Home Forums Rules Articles diyAudio Store Gallery Wiki Blogs Register Donations FAQ Calendar Search Today's Posts Mark Forums Read Search

 Please consider donating to help us continue to serve you. Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
 25th May 2006, 03:05 AM #1 diyAudio Member     Join Date: Oct 2004 Making a milliAmp meter, become a Volt meter I think the answer is on the page linked below, but I can't 'translate' the math, to change my 100mA meter to read volts. The example uses a 1mA meter. Me bad at math http://ourworld.compuserve.com/homep...den/page11.htm =RR=
 25th May 2006, 03:27 AM #2 diyAudio Member   Join Date: Sep 2004 Location: Toronto, Ontario A voltmeter is suppose to have infinite input resistance. AKA no current draw. Getting an ammeter to read volts will be loading the circuit you are trying to measure. Just go and buy the proper voltmeter.
 25th May 2006, 03:29 AM #3 diyAudio Member   Join Date: Jul 2004 Location: nowhere of interest hi redrabbit To answer you we need to know two things. The internal resistance of the meter (just measure it with an ohm meter) which will probably be very small and the number of volts you want for a full scale reading. Simple example (ignoring meter resistance) Say you want to measure up to 100 volts which is handy since you wouldn't have to change the meter scale. R=E/I R=100/0.1 R= 1000 ohms Power rating of resistor, P = I*I*R P=0.1*0.1*1000 P=10 Watts That's a very insensitive meter so it will only be really useful for reading the power supply output voltage. Too much loading for in circuit measuring. Cheers Rob
 25th May 2006, 10:22 AM #4 diyAudio Member   Join Date: Jan 2004 Location: Palmerston North Why not just an opamp set up for voltage controlled current output? Much more elegant solution, probably cheaper than the 10watt resistor and best of all, will hardly load the source. Sam.
 25th May 2006, 10:39 AM #5 diyAudio Member   Join Date: Jul 2004 Location: nowhere of interest Interesting concept. I would like to see the circuit analysis. Given that the op-amp has to source 100 mA, what device did you have in mind? Cheers Rob
 25th May 2006, 11:16 AM #6 diyAudio Member   Join Date: Jan 2004 Location: Palmerston North In simple terms you have the meter connected across Vout to -Vin and then resistor Rf from -Vin to ground. The voltage sensing is from the +Vin of the opamp. The input voltage is directly proportionate to the voltage across Rf developed by the current through the meter and Rf (since the opamp ideally draws no current from -Vin). This solution is also independent of the resistance of the meter. In realistic terms you'd be sampling voltages much higher than the opamp so just add a voltage divider (two resistors) to +Vin. Here's an example: Let's assume the peak voltage is 100V. A voltage divider of 10k and 90k ohms feeds 0 to 10V into +Vin. Rf must match the peak voltage on +Vin when the peak current flows through it, giving: Rf = 10V / 100mA = 100 ohms Now you just have to find a general purpose opamp that will output 100mA. I haven't looked into this myself but I'm sure you can find one. I've attached a pdf from my study guide. Look for the VCIS circuit. There's also an example the of using just a BJT with a few passive components. Doing a discrete solution like the BJT means you can easily get 100mA. Sam.
 25th May 2006, 11:17 AM #7 diyAudio Moderator Emeritus     Join Date: Jan 2003 Location: Near London. UK Forget it. A 100mA meter is completely unsuitable for your purpose. You can pick up a 1mA for peanuts and add a series resistor as rpapps describes to convert it to a 100V meter. The only use for a 100mA meter is... ...measuring current up to 100mA. Edit: Since your 100mA meter is virtually useless, you won't be worried about possibly destroying it. If you open it up from the back I wouldn't be in the least surprised to find that it's actually a 1mA meter with a low resistance shunt resistor across the terminals. If so, it may be possible to convert it into a voltmeter. __________________ The loudspeaker: The only commercial Hi-Fi item where a disproportionate part of the budget isn't spent on the box. And the one where it would make a difference...
 25th May 2006, 01:29 PM #8 On Hiatus     Join Date: Oct 2002 Location: Chicagoland Blog Entries: 2 Two other possibilities: If this meter is salvaged from some bit of equipment, it may just have a 100mA scale, but really be something else. A quick check with an ohmmeter will tell all- sometimes, there's a tiny legend on the scale itself that can tip you off. And on the surplus and salvage market in the US, you're about equally likely to find a 1mA FS and a 50uA FS. If the latter, that's perfect for voltmeter application. __________________ "You tell me whar a man gits his corn pone, en I'll tell you what his 'pinions is."
diyAudio Member

Join Date: Apr 2002
Location: Llanddewi Brefi, NJ
Quote:
 Originally posted by SY Two other possibilities: If this meter is salvaged from some bit of equipment, it may just have a 100mA scale, but really be something else.
100mA meters will almost always have a shunt resistance -- and as SY suggests are 1 ma, 100uA or 50uA full scale. The shunt may be internal in which case you will have to pry the thing apart --

a favorite beginner project is using an MPF102 JFET to drive the meter movement -- this provides a high input impedance -- Ray Marston had one such design on his series in Nuts n Volts -- and thankfully N&V has a free public archive -- the article can be found here: http://www.nutsvolts.com/~downloads/fetjun.pdf

fwiw -- there are some very high quality VOM's (Simpson) and vacuum tube voltmeters (like HP 412's) on the bay --

 25th May 2006, 06:31 PM #10 diyAudio Member   Join Date: Dec 2003 Location: Austin a word of caution about hooking a DVM to a current meter... if it is a low-current meter damage may result from overdriving the motion with the signal from the test equipment (I killed a 10mA meter this way, using the resistance function on a handheld multimeter) __________________ Jesus loves you.

 Posting Rules You may not post new threads You may not post replies You may not post attachments You may not edit your posts BB code is On Smilies are On [IMG] code is On HTML code is OffTrackbacks are Off Pingbacks are Off Refbacks are Off Forum Rules

 Similar Threads Thread Thread Starter Forum Replies Last Post ichiban Parts 8 6th June 2009 03:10 AM ashok Everything Else 3 25th April 2008 03:39 AM Dominick22 Chip Amps 19 24th April 2006 05:58 PM rectaacies Class D 7 3rd February 2006 11:22 AM destroyer X Solid State 10 21st August 2005 05:38 PM

 New To Site? Need Help?

All times are GMT. The time now is 01:21 AM.