
Home  Forums  Rules  Articles  The diyAudio Store  Gallery  Blogs  Register  Donations  FAQ  Calendar  Search  Today's Posts  Mark Forums Read  Search 

Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving 

Thread Tools  Search this Thread 
25th May 2006, 03:05 AM  #1 
diyAudio Member
Join Date: Oct 2004

Making a milliAmp meter, become a Volt meter
I think the answer is on the page linked below, but I can't 'translate' the math, to change my 100mA meter to read volts. The example uses a 1mA meter. Me bad at math
http://ourworld.compuserve.com/homep...den/page11.htm =RR= 
25th May 2006, 03:27 AM  #2 
diyAudio Member
Join Date: Sep 2004
Location: Toronto, Ontario

A voltmeter is suppose to have infinite input resistance. AKA no current draw. Getting an ammeter to read volts will be loading the circuit you are trying to measure. Just go and buy the proper voltmeter.

25th May 2006, 03:29 AM  #3 
diyAudio Member
Join Date: Jul 2004
Location: nowhere of interest

hi redrabbit
To answer you we need to know two things. The internal resistance of the meter (just measure it with an ohm meter) which will probably be very small and the number of volts you want for a full scale reading. Simple example (ignoring meter resistance) Say you want to measure up to 100 volts which is handy since you wouldn't have to change the meter scale. R=E/I R=100/0.1 R= 1000 ohms Power rating of resistor, P = I*I*R P=0.1*0.1*1000 P=10 Watts That's a very insensitive meter so it will only be really useful for reading the power supply output voltage. Too much loading for in circuit measuring. Cheers Rob 
25th May 2006, 10:22 AM  #4 
diyAudio Member

Why not just an opamp set up for voltage controlled current output? Much more elegant solution, probably cheaper than the 10watt resistor and best of all, will hardly load the source.
Sam. 
25th May 2006, 10:39 AM  #5 
diyAudio Member
Join Date: Jul 2004
Location: nowhere of interest

Interesting concept.
I would like to see the circuit analysis. Given that the opamp has to source 100 mA, what device did you have in mind? Cheers Rob 
25th May 2006, 11:16 AM  #6 
diyAudio Member

In simple terms you have the meter connected across Vout to Vin and then resistor Rf from Vin to ground. The voltage sensing is from the +Vin of the opamp.
The input voltage is directly proportionate to the voltage across Rf developed by the current through the meter and Rf (since the opamp ideally draws no current from Vin). This solution is also independent of the resistance of the meter. In realistic terms you'd be sampling voltages much higher than the opamp so just add a voltage divider (two resistors) to +Vin. Here's an example: Let's assume the peak voltage is 100V. A voltage divider of 10k and 90k ohms feeds 0 to 10V into +Vin. Rf must match the peak voltage on +Vin when the peak current flows through it, giving: Rf = 10V / 100mA = 100 ohms Now you just have to find a general purpose opamp that will output 100mA. I haven't looked into this myself but I'm sure you can find one. I've attached a pdf from my study guide. Look for the VCIS circuit. There's also an example the of using just a BJT with a few passive components. Doing a discrete solution like the BJT means you can easily get 100mA. Sam. 
25th May 2006, 11:17 AM  #7 
diyAudio Moderator Emeritus
Join Date: Jan 2003
Location: Near London. UK

Forget it. A 100mA meter is completely unsuitable for your purpose. You can pick up a 1mA for peanuts and add a series resistor as rpapps describes to convert it to a 100V meter. The only use for a 100mA meter is... ...measuring current up to 100mA.
Edit: Since your 100mA meter is virtually useless, you won't be worried about possibly destroying it. If you open it up from the back I wouldn't be in the least surprised to find that it's actually a 1mA meter with a low resistance shunt resistor across the terminals. If so, it may be possible to convert it into a voltmeter.
__________________
The loudspeaker: The only commercial HiFi item where a disproportionate part of the budget isn't spent on the box. And the one where it would make a difference... 
25th May 2006, 01:29 PM  #8 
diyAudio Moderator

Two other possibilities:
If this meter is salvaged from some bit of equipment, it may just have a 100mA scale, but really be something else. A quick check with an ohmmeter will tell all sometimes, there's a tiny legend on the scale itself that can tip you off. And on the surplus and salvage market in the US, you're about equally likely to find a 1mA FS and a 50uA FS. If the latter, that's perfect for voltmeter application.
__________________
"Pity, wrath, heroism, filled them, but the power of putting two and two together was annihilated." EM Forster 
25th May 2006, 01:49 PM  #9  
diyAudio Member

Quote:
a favorite beginner project is using an MPF102 JFET to drive the meter movement  this provides a high input impedance  Ray Marston had one such design on his series in Nuts n Volts  and thankfully N&V has a free public archive  the article can be found here: http://www.nutsvolts.com/~downloads/fetjun.pdf fwiw  there are some very high quality VOM's (Simpson) and vacuum tube voltmeters (like HP 412's) on the bay  

25th May 2006, 06:31 PM  #10 
diyAudio Member
Join Date: Dec 2003
Location: Austin

a word of caution about hooking a DVM to a current meter... if it is a lowcurrent meter damage may result from overdriving the motion with the signal from the test equipment (I killed a 10mA meter this way, using the resistance function on a handheld multimeter)
__________________
Jesus loves you. 
Thread Tools  Search this Thread 


Similar Threads  
Thread  Thread Starter  Forum  Replies  Last Post 
Help! How to measure 2450 volts with 1000 volt max meter.  ichiban  Parts  8  6th June 2009 03:10 AM 
Kenwood VT176 / 175 milli volt meter  ashok  Everything Else  3  25th April 2008 03:39 AM 
Wiring Digital Volt Meter  Dominick22  Chip Amps  19  24th April 2006 05:58 PM 
tamp vu meter and volt meter question....  rectaacies  Class D  7  3rd February 2006 11:22 AM 
I need a portable diy electronic AC volt meter, to avoid transportation of Scopes.  destroyer X  Solid State  10  21st August 2005 05:38 PM 
New To Site?  Need Help? 