An amplifier to test chokes - diyAudio
 An amplifier to test chokes
 User Name Stay logged in? Password
 Home Forums Rules Articles diyAudio Store Gallery Wiki Blogs Register Donations FAQ Calendar Search Today's Posts Mark Forums Read Search

 Please consider donating to help us continue to serve you. Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
diyAudio Member

Join Date: Mar 2003
Location: Zürich
An amplifier to test chokes

Last week I put together this simple Single Ended amplifier to test Power supply chokes.

The choke (DUT) is connected in series with the drain, and parallel to the choke I connected a resistor, with value based on expected inductance at about 100Hz (which is the frequency at which the PS choke will be working). I tested a 159M from Hammond (15H at 100mA), so took a 12k resistor I had on hand.

The big resistor in the source is to further increase the output impedance of the drain. With the potentiometer one can dial in the desired DC current.

I used Steps (Arta) to generate a bode plot (sorry, ommited phase in the graph) to measure the frequency response of this circuit, obtaining the plot below. I fed about 50mV RMS to the gate, output was about 5V RMS (voltage across the choke).

The way I interpret it: the frequency response is relatively flat between 1k and 10k.This is the region where the inductive reactance of the choke is so high that the load to the amp is dominated by the 12k resistor. We are interested in the -3dB points at both extremes: I would say that at about 150Hz we have a -3dB relative to the 1k to 10k band. So at 150Hz the inductive reactance is = load resistor = 12k. Calculating backwards (12000/(2*pi*150)) = L = 12.73H (close to the claimed 15H).

At the top end the -3dB is at about 50kHz, one can calculate capacitance from that.

Is this approach more or less sane? And I was quite impressed that the choke goes up to 20kHz with less than 1dB loss.

Erik
Attached Images
 SingleEndedAmp.jpg (97.2 KB, 111 views) 159m.png (45.6 KB, 105 views)
__________________
my surname is indeed 'de Best': neither misspelling nor snobbism! Ask SY!

 22nd September 2016, 04:31 PM #2 diyAudio Member     Join Date: May 2015 Location: Parana Very clever! Knowing the attenuation points is an interesting tactic, because it will lead to know the real inductance for the application with the custom signal applied. So I will not find strange to find some different values provided by the manufacturer due to differences between methods. But if I make one (and will !), I prefer to use a bigger source resistor, only to be at safe side about the drain resistance. Gary Pimm used one of their modified CCS for measuring a DC power supply (one also based from GPCCS's), injecting variable DC+"AC" (variable DC) current to check load response/output Z etc , so this is a tool for several uses. I've seen a some time ago a guy using a pentode for this approach. Without degeneration, the rp is somewhat low and variable, unless one use a signal pentode like 6AU6 with >1M for ra, but with some heavy cathode resistance the behaviour will be the same (good for tube maniacs like me!). If one needs also to know the low signal response, is possible to use the FR1 or FR2 from ARTA. It uses a pink noise to test, and save some time for measuring. __________________ Making cabinets for OLD projects (better later than never...)
 22nd September 2016, 08:16 PM #3 diyAudio Member   Join Date: Aug 2010 I agree, very clever.
 22nd September 2016, 10:12 PM #4 diyAudio Member     Join Date: Feb 2009 Location: Melbourne, Oz A common assessment is to keep the voltage across the DUT choke a constant, so as to avoid changing the choke's inductance. Did you measure the shunt choke capacitance in the more typical way, and compare results ? Did you measure the non-DC bias inductance using your circuit, and compare that to the typical measurement using just mains frequency applied voltage and a current meter?
 22nd September 2016, 11:13 PM #5 diyAudio Member     Join Date: Apr 2008 Location: Carlisle, England I have designed LLC SMPS and needed to know the leakage inductance of the transformer. I shorted the outputs together. I then applied a sine wave through a known capacitor. I moved the frequency up and down until I got minimum signal across the series LC and this is the resonant frequency. I know the C so its just a matter of re-arranging the formula to work out the L. f=1/(2 pi sqroot(LC) ) __________________ Murton-Pike Systems PCBCAD51 pcb design software. http://www.murtonpikesystems.co.uk
diyAudio Member

Join Date: May 2015
Location: Parana
Quote:
 Originally Posted by nigelwright7557 I have designed LLC SMPS and needed to know the leakage inductance of the transformer. I shorted the outputs together. I then applied a sine wave through a known capacitor. I moved the frequency up and down until I got minimum signal across the series LC and this is the resonant frequency. I know the C so its just a matter of re-arranging the formula to work out the L. f=1/(2 pi sqroot(LC) )
And for audio chokes, we can choose a sufficient large know capacitor to make the resonance to be at 20Hz~100kHz, so we can apply a LIMP like program and found the low/null point in the plotted graph.
__________________
Making cabinets for OLD projects (better later than never...)

 23rd September 2016, 12:01 PM #7 diyAudio Member   Join Date: Mar 2003 Location: Zürich Thanks for all comments! I have not done any other tests, yet - so thanks for your input and suggestions! Cheers, Erik __________________ my surname is indeed 'de Best': neither misspelling nor snobbism! Ask SY!

 Posting Rules You may not post new threads You may not post replies You may not post attachments You may not edit your posts BB code is On Smilies are On [IMG] code is On HTML code is OffTrackbacks are Off Pingbacks are Off Refbacks are Off Forum Rules

 Similar Threads Thread Thread Starter Forum Replies Last Post DIYspaceW Swap Meet 5 13th March 2013 05:10 PM speakerfritz Tubes / Valves 10 20th October 2010 11:32 AM riotubes Parts 0 1st September 2009 04:36 AM audio-gd Solid State 0 4th November 2004 12:07 PM mrfeedback Solid State 17 13th August 2002 02:50 AM

 New To Site? Need Help?

All times are GMT. The time now is 04:28 AM.