|Parts Where to get, and how to make the best bits. PCB's, caps, transformers, etc.|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|13th October 2002, 01:33 AM||#1|
Join Date: Dec 2006
settings for 300b tubes on 539c tester
Can some one please point me in the right direction to find out what the settings for the Hickok 539c vacuum tube tester are for a 300b type tube. I much appreciate this. Thank you.
|19th October 2002, 12:44 PM||#2|
I did a bit of checking and can make a recommendation. I did this by comparing the settings from my TV7 tester and 539C testers. The TV7 tester recommends:
5.0 ER3-2000 60 (bias) C (range) 58 (transconductance)
2.5 ER3-2000 67 (bias) C(range) 38 (transconductance)
On the TV7 a transconductance of 38 (arbitrary units) is actual
1900uMhos. A setting of 58 translates to 2900uMhos
The 539C only has a listing for the 2A3. It recommends:
2.5 ER3-2000 24(bias) D (range) 1900(transconductance)
Extrapolating this to a 300B, I would suggest:
5.0 ER3-2000 21.5(bias) D(range) 2900(transconductance)
Let me know if this works out for you.
|25th February 2014, 06:58 PM||#3|
Join Date: Apr 2008
The 539B settings for the 300A/B tube exists in a factory-issued document called :
MODEL 539B - SUPPLEMENTARY DATA - WESTERN ELECTRIC TUBES
You should be able to find it at :
Hickok Documentation, Manuals, Schematics, and Catalogs
Here's what it states, according to the Hickok 539 roll-chart codification :
Short on A
I am very interested to know your measurements results to compare with the ones I have on my 539B and C...
if it works, don't fix it !
si cela fonctionne, ne touchez à rien !
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|EICO 666 tube tester - What settings do you use?||rotofluid||Tubes / Valves||5||25th July 2012 12:24 PM|
|Precise 111 tube tester - missing tubes||dieter9084d||Tubes / Valves||0||12th July 2009 05:09 PM|
|Looking to "Borrow" an Amplitrex, 539C or other High-End Tube Tester in SE PA Area||baileyler||Swap Meet||0||14th January 2009 03:07 PM|
|Need Tube Testing Settings for 3 tubes||n2lbv||Tubes / Valves||0||29th July 2007 01:37 AM|
|Tube tester settings?||jaymanaa||Tubes / Valves||2||16th August 2006 02:58 AM|
|New To Site?||Need Help?|