Help changing scale of ammeter
 User Name Stay logged in? Password
 Home Forums Rules Articles diyAudio Store Blogs Gallery Wiki Register Donations FAQ Calendar Search Today's Posts Mark Forums Read Search

 Parts Where to get, and how to make the best bits. PCB's, caps, transformers, etc.

 Please consider donating to help us continue to serve you. Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
 12th March 2003, 08:26 PM #1 Sandy H.   diyAudio Member     Join Date: Nov 2002 Location: Charlotte, NC - USA Help changing scale of ammeter While building my bench PSU I found some DC milliammeters for under \$10 and decided to convert them to read the units I was concerned with. The PSU is the Elliot lab supply. Figuring out the size of the series resistor was easy to convert one ammeter to a voltmeter. However, I am not sure I have the correct plan for converting the 0-10 milliammeter to a 0-1 ammeter. Here are my thoughts (per Rod's site and an old physics book): To convert an ammeter to read a different scale, simply determine the current required to move full scale and the meter's resistance. Put the ammeter in parallel with a shunt resistor sized to allow the desired current to flow through the meter. Specifically: (Ifs*Rmeter)/(Iscale-Ifs) where: Ifs is current required for full scale movement of meter Rmeter is the meters resistance Iscale is the desired scale Plugging in values obtained from a gage: Ifs=.01 A Rmeter=240 Ohm and my desired scale, Iscale, of 1 A I obviously get a 2.4 Ohm resistor. Now, running .99 A through a 2.4 Ohm resistor yields a voltage drop nominally of 2.4 V and a power dissipation of nominally 2.4 W. Do these values seem reasonable? I was not planning on having a 2.4 voltage drop per rail due to meters. I also was not intending to dissipate 2.4 W of additional heat through a resistor. These numbers aren't unacceptable, just unexpected. Is it common for an ammeter to cause this magnitude of loss. The example on Rod's website showed an ammeter spec'd at 58 Ohm and .001A full deflection. This obviously results in a tiny shunt resistor (.058 Ohm) and an equally low Voltage drop (.1 V) and very small power consumption. I have not, however, seen any ammeters from places like Mouser or Digikey that are spec'd similarly. Any thoughts, suggestions or additional insight is appreciated. Sandy. PS: I should add that I have not received the actual meters I ordered yet. I was simply wanting to get the calculation format together so when I did, I could get the actual specs and purchase the correct shunt resistor/trimpot.
 12th March 2003, 09:10 PM #2 jackinnj   diyAudio Member     Join Date: Apr 2002 Location: Llanddewi Brefi, NJ you have to determine the internal resistance of the meter, then you can determine the value of the shunt. I put the procedure up at www.tech-diy.com/metershunt.gif
 13th March 2003, 12:04 AM #3 Sandy H.   diyAudio Member     Join Date: Nov 2002 Location: Charlotte, NC - USA Thanks Jack! Thanks for posting the measurement method for determining internal resistance of the meter. I was intending to simply use my multimeter, which the article explicitly states not to do. You saved one of my meters for sure. As to the point of the internal resistance, Rod indicated that for a meter he had on hand, the internal resistance was 58 ohms for full-scale deflection of .001A. Mouser indicated that the internal resistance was around 240 ohms for a 0-10 mA meter. Other similar meters also had a ratio of current flow to resistance which would yield a 2.4 V drop. When the actual meter I purchased arrives, I will use the method you offered to determine what its internal resistance is. I am surprised, though that the analog meters Mouser offers would result in such a high loss. It seems like for many applications, this would be a significant downfall. Are you aware of any technical reason that a 2.4 V loss would make the meter more useful than a much lower loss? Thanks again! Sandy.

 Posting Rules You may not post new threads You may not post replies You may not post attachments You may not edit your posts BB code is On Smilies are On [IMG] code is On HTML code is Off Forum Rules
 Forum Jump User Control Panel Private Messages Subscriptions Who's Online Search Forums Forums Home Site     Site Announcements     Forum Problems Amplifiers     Solid State     Pass Labs     Tubes / Valves     Chip Amps     Class D     Power Supplies     Headphone Systems Source & Line     Analogue Source     Analog Line Level     Digital Source     Digital Line Level     PC Based Loudspeakers     Multi-Way     Full Range     Subwoofers     Planars & Exotics Live Sound     PA Systems     Instruments and Amps Design & Build     Parts     Equipment & Tools     Construction Tips     Software Tools General Interest     Car Audio     diyAudio.com Articles     Music     Everything Else Member Areas     Introductions     The Lounge     Clubs & Events     In Memoriam The Moving Image Commercial Sector     Swap Meet     Group Buys     The diyAudio Store     Vendor Forums         Vendor's Bazaar         Sonic Craft         Apex Jr         Audio Sector         Acoustic Fun         Chipamp         DIY HiFi Supply         Elekit         Elektor         Mains Cables R Us         Parts Connexion         Planet 10 hifi         Quanghao Audio Design         Siliconray Online Electronics Store         Tubelab     Manufacturers         AKSA         Audio Poutine         Musicaltech         Holton Precision Audio         CSS         exaDevices         Feastrex         GedLee         Head 'n' HiFi - Walter         Heatsink USA         miniDSP         SITO Audio         Twin Audio         Twisted Pear         Wild Burro Audio

 Similar Threads Thread Thread Starter Forum Replies Last Post ThSpeakerDude88 Tubes / Valves 15 5th April 2008 02:43 PM dBfreak Swap Meet 1 16th March 2007 10:41 PM xiaonan Parts 6 13th November 2006 07:33 PM Rancid Everything Else 1 19th March 2004 01:01 AM stereth Parts 0 18th October 2003 07:26 PM

 New To Site? Need Help?

All times are GMT. The time now is 05:42 PM.