|Parts Where to get, and how to make the best bits. PCB's, caps, transformers, etc.|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|8th November 2007, 10:00 AM||#1|
Join Date: Jan 2007
Location: Adelaide, South Australia
Tektronix 1101a Psu & P6201 Fet Probe
Just some friendly advice regarding the above power supply.
As some may be aware, one of the best general purpose Fet Probes that can be bought quite cheaply is the P6201 (900 Mhz)
Unfortunately, the matching power supply, the 1101A is still relatively expensive, especially for what it is, ie a PSU that provides +/-15V and +5V
One can eaisily build one quite cheaply, but be aware of a wiring discrepancy in the manual (available from the TEK website)
DO NOT use the wiring diagram schedule on page 1-5. It does not correspond with the pins number designations printed on the LEMO socket.
Pin 3 and 4 have been swapped around.
Use the socket wiring diagram as shown on page 2-4, and ignore the pin numbering. This is the correct one.
Since the probe has no protection against reverse polarity, you can kiss it goodbye. Trust me, I found out the painfull way.
Hope this is of some help to you guys.
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|HV Probe compensation||Elvee||Everything Else||7||23rd November 2007 07:47 AM|
|diy differential probe?||zilog||Class D||7||18th April 2006 09:42 PM|
|DIY HV Probe thisway possible?||Kenshin||Power Supplies||5||9th August 2005 02:35 AM|
|How to compensate a Tek probe?||Bricolo||Parts||0||16th January 2005 12:36 PM|
|DIY oscilloscope probe||Bricolo||Parts||2||25th September 2003 10:56 AM|
|New To Site?||Need Help?|