|11th February 2004, 03:08 AM||#31|
Join Date: Nov 2001
Location: Buenos Aires
I was wondering how difficult would be to mod the meter to add a swich that lets to read 30 db lower (from 20 db to 70db). I need a meter to read residual noise on locations.
I think that providing 30 db more gain on the IC side. may be with another-less noise-OPAMP.
What do you think?
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|Zobel Network Corrections||fotios||Solid State||11||22nd February 2007 06:03 AM|
|how to make this esr meter work with 250uA Panel meter||dytln_02||Everything Else||2||27th January 2007 11:13 PM|
|Rogers Cadet III schematics Corrections||jkeny||Tubes / Valves||4||21st January 2007 09:23 PM|
|Making a milliAmp meter, become a Volt meter||redrabbit||Power Supplies||11||26th May 2006 08:51 PM|
|Impedance Corrections for ESLs||G-Daddy||Planars & Exotics||15||15th November 2001 02:16 PM|
|New To Site?||Need Help?|