Inductance calculator
 User Name Stay logged in? Password
 Home Forums Rules Articles diyAudio Store Blogs Gallery Wiki Register Donations FAQ Calendar Search Today's Posts Mark Forums Read Search

 Everything Else Anything related to audio / video / electronics etc) BUT remember- we have many new forums where your thread may now fit! .... Parts, Equipment & Tools, Construction Tips, Software Tools......

 Please consider donating to help us continue to serve you. Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
Elvee
diyAudio Member

Join Date: Sep 2006
Quote:
 Originally Posted by akis My inductance meter uses frequencies up to 800 KHz. Does the secondary coil affect the inductance of the primary?
If the turn ratio between the secondary and the primary is large, the secondary stray capacitance reflected to the primary will be significant, and if you make the measurement at a high frequency, it will affect the accuracy of the measurement.

Quote:
 If I short the secondary coil I read different values. I think I read somewhere this is called "leaked inductance"?
What you measure is the leakage inductance as seen from the primary.

To make this measurement accurately, your inductance-meter must have a good vectorial discrimination, otherwise the displayed value will be corrupted by the windings resistance.
__________________
. .Circlophone your life !!!! . .
♫♪ My little cheap Circlophoneİ ♫♪

 21st June 2012, 11:36 AM #12 PlasticIsGood   diyAudio Member     Join Date: Mar 2012 Location: Bradford [QUOTE=akis;3054813 The formula says 638 uH but I am measuring 9uH. If I wind about 12 turns and on a slightly larger diameter I measure 739 uH. Is there a way to be able to calculate the inductance without having to wind up and then measure? [/QUOTE] Your calculation shouldn't be so far out. Perhaps the test signal is saturating the core? If that's the case, then inductance will fall dramatically for part of the cycle and put the meter off.
PlasticIsGood
diyAudio Member

Join Date: Mar 2012
Quote:
 Originally Posted by akis My inductance meter uses frequencies up to 800 KHz. Does the secondary coil affect the inductance of the primary? If I short the secondary coil I read different values. I think I read somewhere this is called "leaked inductance"?
You should have said it's a transformer, rather than an inductor, at the start.

Shorting the secondary coil effectively places a resistance across most of the primary inductance, equal to the secondary winding resistance multiplied by square of the turns ratio. Leakage inductance is that part of primary inductance that is not linked to the secondary, so it remains in series.

Basically, your meter is driving current through the secondary and it may not be able to distinguish that from the current through the primary inductance. A transformer is a network of resistors, capacitors and inductors. For accurate results you would need to measure at several frequencies and do some simultaneous equations to eliminate some of the variables.

Leave the secondary open. It then only contributes its winding capacitance.

Current through the secondary makes saturation less likely, by the way.

 22nd June 2012, 06:09 AM #14 akis   diyAudio Member   Join Date: Jul 2008 That means that I can wind just the primary and measure it and there will be no secondary to alter the results with capacitance or leakage inductance ?
Elvee
diyAudio Member

Join Date: Sep 2006
Quote:
 Originally Posted by akis That means that I can wind just the primary and measure it and there will be no secondary to alter the results with capacitance or leakage inductance ?
Yes, if you add the secondary, at best it will change nothing to the measured value, but if the meter is upset by the capacitance, it could display an incorrect value.
Besides, it is easier to correct the number of turns if you don't have to unwind the secondary first.
__________________
. .Circlophone your life !!!! . .
♫♪ My little cheap Circlophoneİ ♫♪

 Posting Rules You may not post new threads You may not post replies You may not post attachments You may not edit your posts BB code is On Smilies are On [IMG] code is On HTML code is Off Forum Rules
 Forum Jump User Control Panel Private Messages Subscriptions Who's Online Search Forums Forums Home Site     Site Announcements     Forum Problems Amplifiers     Solid State     Pass Labs     Tubes / Valves     Chip Amps     Class D     Power Supplies     Headphone Systems Source & Line     Analogue Source     Analog Line Level     Digital Source     Digital Line Level     PC Based Loudspeakers     Multi-Way     Full Range     Subwoofers     Planars & Exotics Live Sound     PA Systems     Instruments and Amps Design & Build     Parts     Equipment & Tools     Construction Tips     Software Tools General Interest     Car Audio     diyAudio.com Articles     Music     Everything Else Member Areas     Introductions     The Lounge     Clubs & Events     In Memoriam The Moving Image Commercial Sector     Swap Meet     Group Buys     The diyAudio Store     Vendor Forums         Vendor's Bazaar         Sonic Craft         Apex Jr         Audio Sector         Acoustic Fun         Chipamp         DIY HiFi Supply         Elekit         Elektor         Mains Cables R Us         Parts Connexion         Planet 10 hifi         Quanghao Audio Design         Siliconray Online Electronics Store         Tubelab     Manufacturers         AKSA         Audio Poutine         Musicaltech         Aussie Amplifiers         CSS         exaDevices         Feastrex         GedLee         Head 'n' HiFi - Walter         Heatsink USA         miniDSP         SITO Audio         Twin Audio         Twisted Pear         Wild Burro Audio

 Similar Threads Thread Thread Starter Forum Replies Last Post patriz Multi-Way 4 29th December 2011 11:11 AM bigpanda Solid State 12 9th April 2008 11:55 AM hitsware Full Range 11 12th November 2004 03:42 PM polsol Multi-Way 5 6th September 2004 07:46 AM

 New To Site? Need Help?

All times are GMT. The time now is 12:28 AM.