Kelvin connection...diy cables
 User Name Stay logged in? Password
 Home Forums Rules Articles diyAudio Store Blogs Gallery Wiki Register Donations FAQ Calendar Search Today's Posts Mark Forums Read Search

 Everything Else Anything related to audio / video / electronics etc) BUT remember- we have many new forums where your thread may now fit! .... Parts, Equipment & Tools, Construction Tips, Software Tools......

 Please consider donating to help us continue to serve you. Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
 7th November 2008, 04:34 PM #1 redrabbit   diyAudio Member     Join Date: Oct 2004 Kelvin connection...diy cables I just got a Valhalla ohm meter (measures <200m to 10.9k ohms). http://www.testequipmentconnection.c...LA_4100ATC.PDF There are Kelvin connections ....4 banana/binding posts....2 for voltage(hi/low), 2 for current (hi/low). When making my own cables, using 2 coax cables, should "current wires be the shield and "voltage" be the inner wire?....or the other way around? Unlike the picture, I'm not using 4 wire cable, I'm using a pair of regular coax. I do have some triax cable somewhere, if adding an extra grounded shield is better. =RR=
 7th November 2008, 05:43 PM #2 redrabbit   diyAudio Member     Join Date: Oct 2004 BTW....searching the net...here might be a DIY low-ohm measuring project for the the inner-nerd inside you. Although I don't know what the best 1k freq source might be..... http://www.analog.com/static/importe...92533AN306.pdf The equation for resistance is : R= 0.0157 x Vout (0.0157 = 15.7mV) ...I wonder is a DC volt panel meter could be implemented to calculate the equation, to display the actual ohms...?? =FB=
 7th November 2008, 06:04 PM #3 Iain McNeill   diyAudio Moderator Emeritus     Join Date: Oct 2007 Location: Santa Cruz, California The Kelvin connection is to eliminate the volt drop due to the current flowing in the test leads affecting the voltage measurement. The voltmeter is high impedance and so no current flows in these leads. The voltage at the meter is thus the same as the voltage across the resistor under test. As the voltmeter is high impedance I would use a shielded cable for the voltage measurement and a second cable for the current so that the current doesn't inductively couple into the voltage reading. If you have a good constant current source then a DPM can be calibrated to read ohms directly over a limited range. Switching the current will extend the range.

 Posting Rules You may not post new threads You may not post replies You may not post attachments You may not edit your posts BB code is On Smilies are On [IMG] code is On HTML code is Off Forum Rules
 Forum Jump User Control Panel Private Messages Subscriptions Who's Online Search Forums Forums Home Site     Site Announcements     Forum Problems Amplifiers     Solid State     Pass Labs     Tubes / Valves     Chip Amps     Class D     Power Supplies     Headphone Systems Source & Line     Analogue Source     Analog Line Level     Digital Source     Digital Line Level     PC Based Loudspeakers     Multi-Way     Full Range     Subwoofers     Planars & Exotics Live Sound     PA Systems     Instruments and Amps Design & Build     Parts     Equipment & Tools     Construction Tips     Software Tools General Interest     Car Audio     diyAudio.com Articles     Music     Everything Else Member Areas     Introductions     The Lounge     Clubs & Events     In Memoriam The Moving Image Commercial Sector     Swap Meet     Group Buys     The diyAudio Store     Vendor Forums         Vendor's Bazaar         Sonic Craft         Apex Jr         Audio Sector         Acoustic Fun         Chipamp         DIY HiFi Supply         Elekit         Elektor         Mains Cables R Us         Parts Connexion         Planet 10 hifi         Quanghao Audio Design         Siliconray Online Electronics Store         Tubelab     Manufacturers         AKSA         Audio Poutine         Musicaltech         Aussie Amplifiers         CSS         exaDevices         Feastrex         GedLee         Head 'n' HiFi - Walter         Heatsink USA         miniDSP         SITO Audio         Twin Audio         Twisted Pear         Wild Burro Audio

 Similar Threads Thread Thread Starter Forum Replies Last Post forart.eu Parts 12 23rd May 2009 02:15 PM waynzworld Solid State 1 8th January 2007 03:14 PM johnferrier Everything Else 3 18th October 2003 02:07 PM

 New To Site? Need Help?

All times are GMT. The time now is 08:41 PM.