|Equipment & Tools From test equipment to hand tools|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|16th December 2005, 04:55 PM||#1|
Join Date: Feb 2004
TIM-measuring with RE201 Analyzer
is there anybody using a RE-201 audio analyzer?
The internal TIM-generator is fine and supplies the TIM signal without any problems. The datasheet of the RE201 promise a distortion of 0,02% for the TIM generator itself.
The Tim-signal is shown absolute "clear" on a spectrum analyzer down to -80 dB....no really visible TIM components to see.
The "Problem" is the TIM analyzer. Specifications of the analyzer tell me a sensitivity of only 0,2% (-55 dB)
When i connect the generator directly to the analyzer, a value of 0,14 % TIM will be displayed.
But this value doesn´t match the display on the FFT Analyzer.
i ask myself if this is ok??
Anyway....a (weak) sensitivity of 0,2% is far away from practice, and the same RE 201 can measure THD down to 0,0004%
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|How to set up FFT Analyzer||Sunsun22||Solid State||1||15th April 2009 04:49 PM|
|FFT Analyzer||Radioman62||Parts||0||1st April 2009 09:13 PM|
|FFT analyzer for MLS testing||sonicman||Multi-Way||0||21st September 2008 12:37 AM|
|UPA audio-analyzer||azaz||Swap Meet||0||13th August 2007 10:00 AM|
|Audio Analyzer||audioPT||Everything Else||0||13th August 2003 11:12 PM|
|New To Site?||Need Help?|