HP 1821a issues
the 1821a double time base in my old HP 184A oscilloscope mainframe has got some issues. Switched to three of all the sweep frequencies, the beam doesn't start at the left edge of the screen, but about 1 cm right from it. The trouble shooting scheme at page 50 of the related manual tells me to check the transistors Q210 and Q211 and CKTS. What does CKTS mean?
'CKTS" I would assume is short for "circuits". I guess they mean look at the circuitry involving Q210 and Q211. I used to fix HP analog scopes way back and they were a royal pain to fix. HP analog scopes had lots of issues!!
|All times are GMT. The time now is 07:33 PM.|
vBulletin Optimisation provided by vB Optimise (Pro) - vBulletin Mods & Addons Copyright © 2016 DragonByte Technologies Ltd.
Copyright ©1999-2016 diyAudio