|Equipment & Tools From test equipment to hand tools|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|3rd September 2003, 05:35 AM||#1|
Hitachi V202 oscilloscope issues
I have an old Hitachi V-202 20 Mhz oscilloscope.
During the rainy season we had a really bad day and some moisture condensed on the components. When I turned it on it blew some TO220 type power transistors on the main board. I replaced these. The display however seemed to loose intensity control . So I had to solder a 3.3M resistor in series with the control grid ( I think) and it now has a fixed acceptable brightness.
All other parts 'seem' to be OK. Does anyone have any idea what is happening ? I have a circuit diagram of the intensity control circuit which I will scan and post later. I did see a flash around the connector of the CRT when I opened it and turned it on ( at the time it failed). That must be responsible for the blown power transistors.
I am able to use the unit now though the intensity is fixed. It also lost its ability to have an external (Z) intensity control.
It used to be a good scope - still is ! I wish I could fix it again.
The slide switches for the ac/gnd/dc are very hard to operate and have poor contacts. Can I use any switch or should I get them as spares from Hitachi ?
What is the best contact cleaner that I can use ?
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|Hitachi HA5300 ?||Zen Mod||Solid State||31||6th June 2014 07:20 PM|
|Hitachi Cap??||peter7||Pass Labs||4||20th April 2011 10:37 PM|
|hitachi ha-330 modification||adason||Solid State||44||29th March 2010 07:59 AM|
|Crazy issues with an Heathkit oscilloscope||00940||Equipment & Tools||2||5th May 2009 11:47 PM|
|Hitachi D-E66||GW4JUW||Analogue Source||1||4th September 2005 07:44 AM|
|New To Site?||Need Help?|