|Equipment & Tools From test equipment to hand tools|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|8th October 2010, 08:07 AM||#1|
Join Date: Oct 2009
Tektronix 468 horizontal problem in storage mode
I´m having some trouble with my Tektronix 468 and I would be very gratefull if you could give me some advice.
In real mode it works perfectly (even the uncal knob and light), but in storage mode it goes crazy.
Here are the symptoms in storage mode:
-The time/div is not correct (it is about 20 times what it should be, so the signal looks 20 times wider)
-The wave displayed is distorted (square wave looks like a rounded trapezoid)
-The display jumps sideways a bit.
-The uncal lamp, when on, is very dark
-The uncal knob when turned on does nothing
-There is a vertical line at the end of the beam with 2 dots on the edges (see pic)
On both input channels the results are the same, and the vertical circuits and trigger circuits are both working fine, so I guess it is something on the top board, the storage part?
Do you have any idea of what this could be?
|19th October 2010, 12:59 AM||#3|
Join Date: Jan 2010
Location: Jeffersonville, Indiana USA
Bought a Tek 468 25 years ago at work - thank you Generous Elec. Yours has to be nearly as old and expensive electrolytic capacitors last - - - 20 years. My favorite rant, you should stop thinking, look at the part list, and buy one of everything electrolytic. Discharge caps with resistor , measure below 25 VDC before touching - lethal voltages in there. Change two caps at a time and then test, since you might mess up or the new ones might be unsuitable in some way. If doesn't work as least as well as before the latest caps went in- back up with the old ones and think a bit about the circuit.
If you want it to stay fixed, use caps that are rated 2000 hours or more by the manufacturer. Newark.com or farnell.co.uk have the hours shown in their selector tables for caps. Don't buy sale caps, they are going out of date already.
Left that job 15 years ago, now making do with a 20 MHz B&K non- memory scope. Oh, well, back to reality as the song goes.
Dynakit ST70, ST120, PAS2,Hammond H182(2 ea),H112,A100,10-82TC,Peavey CS800S,1.3K, SP2-XT's, T-300 HF Proj's, Steinway console, Herald RA88a mixer, Wurlitzer 4500, 4300
Last edited by indianajo; 19th October 2010 at 01:25 AM.
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|Itu-r 468||gbmaizol||Class D||0||16th August 2010 07:33 AM|
|PCX1250 2ohm mode Problem||ctg353535||Car Audio||1||15th April 2007 03:26 AM|
|Tektronix 2235 problem||Dhul||Parts||8||15th January 2007 01:28 AM|
|Using a spdt switch to select between triode mode and pentode mode on a guitar amp.||G||Tubes / Valves||19||15th March 2006 10:15 AM|
|Calculating watt requirements for multi-mode/tri-mode operation?||Datoyminaytah||Solid State||1||14th August 2004 08:43 AM|
|New To Site?||Need Help?|