|Equipment & Tools From test equipment to hand tools|
Please consider donating to help us continue to serve you.
Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
||Thread Tools||Search this Thread|
|18th June 2010, 06:24 PM||#1|
Join Date: Dec 2005
Location: Brooklyn, New York
Best LDR for Sound Technology ST 17xx gear
Since there is little mentioned on the 'net re Sound Technology gear, perhaps it would be a good idea for those who have successfully repaired their gear to list the LDRs used along with any mods they have made to the circuit(s). Both in the Oscillator & Analyzer sections.
As a folllow up, they can comment on their findings. In addition, it would be helpful to establish a listing of results using different LDRs in different locations on the analyzer board.
U7 (OSCILLATOR board)
The 17xx models share similar LDR circuits.
Last edited by jblmar; 18th June 2010 at 06:32 PM.
|28th June 2010, 12:23 AM||#2|
Join Date: Mar 2004
Location: SF Bay Area
Blog Entries: 2
This is copied from another reply by me. I'm adding it here so I can find it myself later:
When it switches from fast to low distortion it engages the LDR. Those are the real weak point in the analyzer (aside from the many switches). ST didn't reveal the actual part number. The original vendor doesn't exist. Vactec does : Welcome | PerkinElmer so if we can figure out which part it was we can cross check- Silonex Inc.: Products: Audiohm Optocouplers . My schematic shows a CL614m note next to the part. That seems to lead to a black hole. I have a very foggy memory of 200 Ohms on resistance. The coupler has a 10K parallel resistance so that may be a real target value. I believe a VTL5C10 will work as a replacement so the Audiohm http://www.silonex.com/datasheets/sp...pdf/103465.pdf NSL-32B-101 might be a good replacement.
Product Design Services
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|Sound Technology 1710A (Balanced)||Michael Rothacher||Swap Meet||4||24th November 2010 01:00 PM|
|Sound Technology ST 1701A Not Settling In Distortion Mode||jblmar||Equipment & Tools||22||15th June 2010 01:49 AM|
|Sound Technology still in business?||johngalt47||Equipment & Tools||2||18th May 2010 10:32 AM|
|Speaker cable for Definetive Technology ST||jzvjg||Everything Else||0||16th February 2010 05:58 PM|
|Sound Technology ST 1000A & 1701A||jblmar||Solid State||11||5th March 2007 03:31 PM|
|New To Site?||Need Help?|