|24th November 2009, 06:59 AM||#11|
Join Date: Nov 2009
Hello all--newcomer here,
I just purchased a Peak ESR70 a couple weeks ago, in deference to the "blue" kit, and found this thread while searching for more info. It works quite nicely however I found the leads to be absurdly short and the supplied crocodile (alligator here in Florida) clips to be worthless for in-circuit testing.
The lead length was an easy fix, I cut the twin coax line off of a cheap set of "ear-buds" and used both the shield and center of each cable for each pole of the tester's input.
To address the in-circuit issue I made up a set of tweezers from Popsicle sticks--nothing magical in the construction, just cut the pieces/parts to fit and glue 'em up--I built up each leaf (tweeze?) with stacked pieces to provide a grip and then applied the heat-shrink tubing.
The probe tips are 1" brass escutcheon pins bent up as needed and epoxied in place--I have since bent the probe tips inward so as to grab solder pads more forcefully. In use I clip the croc's to the exposed head of the pins.
It all works well and will calibrate to 0.00Ω with no problems...
|Thread Tools||Search this Thread|
|Thread||Thread Starter||Forum||Replies||Last Post|
|My VU meter||gev||Analog Line Level||1||11th April 2007 08:21 AM|
|how to make this esr meter work with 250uA Panel meter||dytln_02||Everything Else||2||27th January 2007 11:13 PM|
|inductance meter from a cheap meter?||jarthel||Parts||3||22nd September 2006 06:37 AM|
|Making a milliAmp meter, become a Volt meter||redrabbit||Power Supplies||11||26th May 2006 08:51 PM|
|t-amp vu meter and volt meter question....||rectaacies||Class D||7||3rd February 2006 12:22 PM|
|New To Site?||Need Help?|