digital scopes and distortion measurement
 User Name Stay logged in? Password
 Home Forums Rules Articles diyAudio Store Blogs Gallery Wiki Register Donations FAQ Calendar Search Today's Posts Mark Forums Read Search

 Equipment & Tools From test equipment to hand tools

 Please consider donating to help us continue to serve you. Ads on/off / Custom Title / More PMs / More album space / Advanced printing & mass image saving
 12th October 2012, 09:42 AM #1 akis   diyAudio Member   Join Date: Jul 2008 digital scopes and distortion measurement There is a long thread in this forum, discussing ultra low distortion audio range oscillators. It is mentioned in some posts that one purpose of an ultra low audio-range oscillator would be to be used in a distortion analyser bench top instrument. I have no experience on the subject but I have bought a digital scope which made me think. The digital scope samples the incoming wave and records the amplitude of the wave over time. Surely then it is simply a matter of comparing these sample points to their mathematically sinusoidal equivalent, ie you do not need a perfect oscillator to make a comparison, you can simply derive the theoretical point on the curve mathematically. I am sure my scope can export tables of data so then I could do that in Excel, even if my scope does not offer a "THD" function out of the box (which I am very puzzled if it does and if not, then why not!). Am I thinking on the right track?
 12th October 2012, 10:17 AM #2 demeterart   diyAudio Member     Join Date: Jan 2012 Location: Athens, Greece digital oscilloscopes can often compute the DFT of the sampled signal and thus allow you to deduce the fidelity of your oscillator compared to a reference sinusoid.
 12th October 2012, 10:21 AM #3 Samuel Groner   diyAudio Member   Join Date: Jun 2005 Location: Zürich Basically yes, but the frontend and converter of your scope will add its own distortion. Probably no better than -80 dB, so not much use for audio... Samuel
macboy
diyAudio Member

Join Date: Oct 2003
Quote:
 Originally Posted by akis There is a long thread in this forum, discussing ultra low distortion audio range oscillators. It is mentioned in some posts that one purpose of an ultra low audio-range oscillator would be to be used in a distortion analyser bench top instrument. I have no experience on the subject but I have bought a digital scope which made me think. The digital scope samples the incoming wave and records the amplitude of the wave over time. Surely then it is simply a matter of comparing these sample points to their mathematically sinusoidal equivalent, ie you do not need a perfect oscillator to make a comparison, you can simply derive the theoretical point on the curve mathematically.
Think about that carefully... compare what to what? So you are thinking that the scope can capture the waveform from the DUT (an amp or whatever) and compare the waveform to "perfect" sine to determine THD? Where does the waveform from the amp come from in the first place? You need an oscillator, as close to perfect (low distortion) as possible, to source the signal that will go through the DUT. If the oscillator is clean enough, then the distortion measured is essentially all contributed by the DUT.

On the measuring end, there are a couple ways to do it, you can examine the spectra of the signal using FFT or a specturm analyzer (essentially a swept bandpass filter), and/or you can pass the signal through a notch filter to remove "all" of the fundamental, leaving only distortion and noise, and measure/observe that.

 12th October 2012, 06:21 PM #5 jcx   diyAudio Member   Join Date: Feb 2003 Location: .. classic Tektronix, HP/Aligent, similar benchtop digital scopes are relatively low res by audio distortion measurement standards many have 8 bit flash converters (but sample rates up to GHz) which may achieve 10 bit linearity if you average this limits their distortion measurement to 50-60 dB while decent soundcards do better than -100 dB thd == over 16 bits linearity and SOTA monlithic ADC can push -120 thd in audio bandwidth Last edited by jcx; 12th October 2012 at 06:27 PM.
 12th October 2012, 06:26 PM #6 PChi   diyAudio Member   Join Date: Jan 2012 Location: Staines, UK I guess in theory yes but in practice no. An Oscilloscope Analog to Digital converter has a fast sampling rate but a low resolution, typically 8 bit. The FFT functions that they offer is limited by the Analog to Digtal Converter resolution. I guess that a sound card could be used but identifying whether the distortion came from the source or was caused by the unit under test seems difficult.

 Posting Rules You may not post new threads You may not post replies You may not post attachments You may not edit your posts BB code is On Smilies are On [IMG] code is On HTML code is Off Forum Rules
 Forum Jump User Control Panel Private Messages Subscriptions Who's Online Search Forums Forums Home Site     Site Announcements     Forum Problems Amplifiers     Solid State     Pass Labs     Tubes / Valves     Chip Amps     Class D     Power Supplies     Headphone Systems Source & Line     Analogue Source     Analog Line Level     Digital Source     Digital Line Level     PC Based Loudspeakers     Multi-Way     Full Range     Subwoofers     Planars & Exotics Live Sound     PA Systems     Instruments and Amps Design & Build     Parts     Equipment & Tools     Construction Tips     Software Tools General Interest     Car Audio     diyAudio.com Articles     Music     Everything Else Member Areas     Introductions     The Lounge     Clubs & Events     In Memoriam The Moving Image Commercial Sector     Swap Meet     Group Buys     The diyAudio Store     Vendor Forums         Vendor's Bazaar         Sonic Craft         Apex Jr         Audio Sector         Acoustic Fun         Chipamp         DIY HiFi Supply         Elekit         Elektor         Mains Cables R Us         Parts Connexion         Planet 10 hifi         Quanghao Audio Design         Siliconray Online Electronics Store         Tubelab     Manufacturers         AKSA         Audio Poutine         Musicaltech         Aussie Amplifiers         CSS         exaDevices         Feastrex         GedLee         Head 'n' HiFi - Walter         Heatsink USA         miniDSP         SITO Audio         Twin Audio         Twisted Pear         Wild Burro Audio

 Similar Threads Thread Thread Starter Forum Replies Last Post Loudthud Equipment & Tools 3 15th August 2012 10:28 PM peranders Software Tools 8 19th April 2010 08:04 PM PMA Multi-Way 15 3rd November 2009 06:08 AM djdamix Solid State 1 20th August 2004 07:11 AM audiobot Tubes / Valves 9 23rd January 2004 12:14 PM

 New To Site? Need Help?

All times are GMT. The time now is 09:12 PM.